Handheld Flow Meter
- Accuracy:Better than ±1%
- Repeatability:Better than 0.2%
- Medium Temperature:-30℃~160℃
- linearity:Better than 0.5%
V-method installation is the moswidely used mode for daily measurement with pipe inner diameters ranging from 15 millimeter to 200 millimeter. It is also called reflective mode or method.
Z-method is commonly used when the pipe diameter is between 300 millimeters and 500 millimeters.
W-method is usually used on plastic pipes with a diameter from 15 millimeters to 100 millimeters
Handheld Ultrasonic Flow Meter Specification:
|Accuracy||±1% of reading at rates>0.2mps|
|Rate Units||Meter,Feet,Cubic Meter,Liter,Cubic Feet,USA Gallon,UK Gallon,
Oil Barrel,USA Liquid Barrel, UK Liquid Barrel;
|Totalizer||7-digit totalizers for net accumulation and positive and negative accumulations|
|Liquid Types||all liquids|
|Security||Locks of Set values, you have to unlock while changing data;|
|Display||4x8 Chinese or 4x16 English letters|
|Communication Interface||RS-232C,baud-rate :from 75 to 57600.Protocol made by the
manufacturer and compatible with that of the FUJI ultrasonic flow meter.
|Transducers||Model M1 for standard,other 3 models for optional|
|Transducer Cord||Standard 2x5 meters,optional 2x10 meters|
|Power Supply||3AAA Ni-Hbuilt-inbatteries.When fully rech arged it will lastover 10 hours of operation.|
|100V-240VAC for the charger|
|Data Logger||Built-in data logger can store over 2000 lines of data|
|Manual||7-digit press-key-to-gototalizer for calibration|